Ok, finally I found the Problem.
I changed the prescaler value of Timer0 becuase I needed a higher PWM Frequenciy.
Unfortuneatly this also affects the millis() function that is used in gw.wait.
It turns out taht with the changed TImer 0 Prescaler 1ms was not 1ms anymore but less. So the gw.wait(750) where only 750ms/64 which is not enough for DS18S20 conversion to take place.
I am non technical but I have 8 wirsbo stats in my house. One of them is faulty and the unit is now obsolete. Is it possible to purchase one of these pcbs and use it to replace the one in the faulty stat ?
Hi @berkseo. Thanks for the suggestion, but could you describe in more detail? As for my TODO list, there is already a task to add 485 interface to the gateway. My latest device "xRoom" have this interface.
ps/ Where have you gone?
I have been working on "xRoom" shield. A small announce:
Do you remember the discussion about multinode? So, it will be available soon))
Wow, great! Its alway nice to see others using my work too. Your code improvement seem to be very helpful too.
I have most recently also integrated this directly into my gateway build (see here). I am still struggling with code there though.
Sorry, 1 more question. At what timeframe do you expect diy packages to be for sale? I have checked with steel manufactures before for other projects, but having a small batch of steel tubes means that they will be enormously expensive and the same goes for the PCB's. The milling I can do easily at a maker space (fablab013). I am planning on a rather long, 1 storey hive, so I plan to cut the metal tubes in half and mount an extension in between.